A Complete HIOKI Digital & Analog Clamp Tester /A AC .. CARRYING CASE ×1. Hand strap ×1. Instruction manual ×1. Hioki Pdf User Manuals. View online or download Hioki Instruction Manual. View and Download Hioki instruction manual online. Clamp On AC/DC HiTester. Measuring Instruments pdf manual download.
|Published (Last):||1 February 2018|
|PDF File Size:||15.32 Mb|
|ePub File Size:||19.54 Mb|
|Price:||Free* [*Free Regsitration Required]|
Remove the rear cover. We accept these payment methods: Next, insert the red test lead to V and the black test lead to COM of the voltage measurement terminal. Save my login information. The counter update may be altered according to measurement conditions. Chapter 2 Measurement Procedure Explains how to use the for measurement. Measuring the Starting Current. Replace the battery hiiki ensure accuracy.
However, mishandling during use could result in injury or death, as well as damage to the instrument. Select a proper mode according to the waveform shown below: There isn’t an annunciator ” “. Some of our pricing is already discounted.
The torque about 0. When all segments on the bar graph go on, the elapsed time is 30 minutes.
Unit turns off during use. The measurement mode is switched as follows. Set a range based on the unit’s measurement range and other instruments, such as recorders. Figs 2 and 3 As the measured value is small to the range response time becomes long.
Not finding what you’re looking for? If the function is activated when there is no input, the minimum value will remain zero. While the recording function is being deactivated, data is not updated, even if the clamp sensor is disconnected from the conductor. Refer to “Chapter4 Battery Replacement”.
Hioki 3284 Instruction Manual
Press to turn the unit on. If the unit is damaged, or fails to operate according to the specifications, contact your dealer or HIOKI representative. HIOKI cannot bear any responsibility for damage that occurs during shipment. It is recommended that a new battery should be used, when the manua, seems to be short. In the manual, the symbol indicates particularly important information that the user should read before using the instrument.
These are defined as follows.
AC/DC Clamp Meter | Clamp On AC/DC HiTester – Hioki
Open the top ends of the clamp core and clamp the measured conductor so that it passes through the center of the clamp core. Ask our experts To ask our Hioki experts a question please include your contact details.
Using a one-chip microprocessor, the Call us toll free 1. In particular, check the accessories, panel switches, keys, and terminals. Then the model name is shown, and the bar graph indicates the battery condition. Pack the unit carefully so that it will not be damaged during transport, and write a detailed description of the problem. If the current range of the measured circuit is known, set the current range to the manual range.
Inrush current crest value. Also, when deactivating the recording function, press the key to terminate HOLD Shipping Check In particular, check the accessories, panel switches, keys, and terminals. By pressing ‘print’ button you will print only current page. FAST Instead, the unit symbol blinks. We have tried to bring this manual as close to perfection as we could achieve.
Press and select a measurement mode for the measured circuit. A conversion table for ranges is provided below. Ask an Expert More Hioki Products. We also accept wire transfers, cheques and purchase orders on approved credit. Also See for Instruction manual – pages. Don’t show me this message again.